Keep batteries out of reach of children Choking hazard. Small parts. Keep away from children under the age of 3 years. Keep new and used batteries away from children.
|Published (Last):||18 January 2017|
|PDF File Size:||14.87 Mb|
|ePub File Size:||1.77 Mb|
|Price:||Free* [*Free Regsitration Required]|
Canada Canada This website contains information on products which are targeted to a wide range of audiences and could contain product details or information otherwise not accessible or valid in your country. Please be aware that we do not take any responsibility for accessing such information which may not comply with any valid legal process, regulation, registration or usage in the country of your origin.
All rights reserved. The test strip may be damaged or not properly inserted. Remove and reinsert the test strip, or replace it if damaged. Your blood glucose may be extremely high or a meter or a test strip error has occurred.
If your test result matches how you feel, contact your healthcare professional immediately. If your test result does not match how you feel, repeat the blood glucose test. Contact your healthcare professional immediately. If the second test result does not match how you feel, perform a control test with the control solution and a new test strip. If the control result is within the acceptable range, review the proper testing procedure and repeat the blood glucose test with a new test strip.
If the control result is not within the acceptable range, contact Accu-Chek Customer Care at Not enough blood or control solution was drawn into the test strip for measurement or was applied after the test had started. Discard the test strip and repeat the blood glucose or control test. Blood or control solution was applied to the test strip before the flashing drop symbol appeared on the display. An electronic error occurred, or in rare cases, a used test strip was removed and reinserted.
Turn the meter on and off, or take the battery out for 20 seconds and reinsert it. Perform a blood glucose or control test. The temperature is above or below the proper range for the system. Refer to the test strip package insert for system operating conditions.
Do not artificially heat or cool the meter. The battery is almost out of power. Change the battery now. If the message reappears after the battery has been replaced, remove the battery again, press any meter button, then reinsert the battery.
The time and date settings may be incorrect. Make sure the time and date are correct and adjust, if necessary. The meter is ready for you to insert a test strip. The meter is ready for a drop of blood or control solution. Your meter measures blood glucose within a defined range 0. The display may indicate that your blood glucose is above the measuring range of the system Immediately repeat the test. If this does not match how you feel, contact Accu-Chek Customer Carer at The display may indicate that your blood glucose is lower than the measuring range of the system 0.
Battery is dead. Insert new battery. Insert one 3-volt lithium battery coin cell type CR Display is damaged. Contact Accu-Chek Customer Carer at Meter is defective.
Extreme temperatures. Move the meter to a more temperate area. The meter is in set-up mode, waiting for you to change or confirm settings.
Accu-Chek Aviva User Manual