Page Site location No. Setting The site location where measurements done can be saved with numbers. Page 23 Time Setting can be done in following procedure. The setting of Rk can be done in following procedure.
|Published (Last):||18 October 2018|
|PDF File Size:||2.15 Mb|
|ePub File Size:||14.13 Mb|
|Price:||Free* [*Free Regsitration Required]|
Page Site location No. Setting The site location where measurements done can be saved with numbers. Page 23 Time Setting can be done in following procedure. The setting of Rk can be done in following procedure. Page Backlight Fig. Page Series Interference Voltage earth Voltage Measurement Series Interference Voltage Earth Voltage Measurement Measurements are automatically started during a measurement of earth resistances and earth resistivities, and the results can be checked on the Result display screen.
Page Measurement Method 9. When connecting the Probes, they should be separated. The measured earth resistances are free of auxiliary earth resistances but the resistances on the E terminal are contained. Firmly insert each plug of 4 Test Leads green, black, yellow, red , to the corresponding connectors on the instrument.
Recheck the connection of Test Leads. Page 36 e. The spike should be stuck in the depth of 25cm or less when the interval of the Auxiliary Earth Spikes is 5m. If the Spikes stuck too deep, it may result in inaccurate earth resistivity measurement. In this case, rotate the Range Switch and select an upper Range.
Kyoritsu 4106 Ground Resistance / Earth Resistivity Tester
Medidor de Tierras y Resistividad con comunicacion a PC
KYORITSU KEW 4106 Instruction Manual
A manufacturer developing, producing and selling electrical measuring equipment